FAQ/009 KAM-500 - KAD/ADC/109/S2: I’m using the KAD/ADC/109/B/S2 module. If I do not want to use the sense line, what do you recommend?
August 18, 2014
Connect the sense line + to the excitation + (respectively sense line - to the excitation -) with a short or tick cable. Sense lines are for compensating voltage drop over excitation lines (cable resistance). Those drops causes gain error of measurement. But with short or/and thick cables, this should be negligible.